DESIGNED FOR PENTAX – One product range that fits every PENTAX Medical endoscope
Manually pre-cleaning flexible endoscopes is key to ensuring reliable reprocessing as an end result.
Even with magnification, visual assessment of the manual pre-cleaning results is insufficient to detect microorganisms.
As such, PENTAX Medical provides a verified and validated solution to improve confidence in manual pre-cleaning, with the new PROfILE single-use cleaning brushes.
- Proven cleaning efficiency
- Designed and validated to meet PENTAX Medical’s quality standards
- Compliant with the Industry’s highest reprocessing standards
To learn more, please visit www.the-hygiene-solution-that-fits.com
Proven cleaning efficiency
In combination with PENTAX Medical endoscopes, the PROfILE single-use cleaning brushes were tested against residual soil and compared against the performance of legacy brushes. Through an authorized statistical test method, the internal validation proved both brushes to have a similar performance.
Designed and validated to meet PENTAX Medical's quality standards
PROfILE single-use cleaning brushes were made to meet the high-quality standards of PENTAX Medical engineering. Extensive internal tests have proven that they are the best possible fit to ensure proper hygiene results, with no impairment to the endoscope, across all types.
- Tailored for PENTAX Medical endoscopes' precise dimensions, structures, and materials
- Improved bonding and pull-off capability at optimal insertion resistance of ≤10N
- No damage to endoscope working channels and reliable even at increased thermal conditions
Compliant with the industry's highest reprocessing standards
How clean is clean enough? This was the leading question of PENTAX Medical engineers when deciding upon the validation criteria for hygienic efficacy of the PROfILE single-use cleaning brushes. In order to follow established standards, PENTAX Medical validated against the requirements of the Association for the Advancement of Medical Instrumentation (AAMI)1.
|Hemoglobin ( <2.2μg/cm2 ) ||Protein ( <6.4μg/cm2 ) ||Carbohydrate (<1.8μg/cm2 ) |
Test example: Residual of three makers at scope's distal end after third cleaning cycle, complying with AAMI requirements.
1 AAMI TIR30:2011 Hemoglobin < 2.2μg/cm2, Protein < 6.4μg/cm2, Carbohydrate < 1.8μg/cm2
AAMI standards are not applicable for Europe.
|Long channel brushes|
|300600 / CS2010A|
|300601 / CS2037A|
|300602 / CS3022A|
|300603 / CS4010A|
|300604 / CS5503A|
|300605 / CS5522A|
|300606 / CS5530A|
|Short inlet seal brushes|
|300607 / CS-C11A|
|300608 / CS-C13A|
|Length||Outer Diameter||Color||Compatible channel width||Working Length||Compatible Endoscopes|
|Gastroenterology||Pulmonology||Urology||ENT & Speech||Intubation|
|(mm)||(mm)||(Shaft / Handle)||(mm)||(mm)|
|1,020||2||Blue||1.1 - 1.5||Up to 720||--||Yes||Yes||Yes||Yes|
|3,720||2||Blue||1.1 - 1.5||Up to 3,420||Yes||--||--||--||--|
|2,220||3||White||1.7 - 2.4||Up to 1,920||Yes||--||--||--||--|
|1,020||4||Orange||1.8 - 3.2||Up to 720||--||Yes||Yes||Yes||Yes|
|310||5,5||Yellow||2.0 - 5.0||N/A||--||Yes||Yes||Yes||Yes|
|2,420||5,5||Yellow||2.0 - 5.0||Up to 2,120||Yes||--||--||--||--|
|3,020||5,5||Yellow||2.0 - 5.0||Up to 2,720||Yes||--||--||--||--|
Please click on product model to view related accessories